Clear View Designs, Inc.
a technology concern
Using built-in RFNoiseCheck software on a Siemens 3-Tesla "Prisma" scanner, the panel to the left shows the baseline RF noise spanning the imaging frequency band. The panel to the right shows the noise with MREyes in operation.
The fact that these noise tests do not differ indicates that MREyes adds no interference to images obtained on the scanner.